If you still have the install disc , you can also partition your primary and install W XP on the new partition from http: At the A site, log a 4A or 3A sequence depending on the size of the clearing. Transfer the file to FV Procedure 3. Synchronize clocks in both wands Procedure 1. AmazonGlobal Ship Orders Internationally. Click OK when finished. Tell us some more!
|Date Added:||24 April 2015|
|File Size:||24.69 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
Take the below-canopy readings along a line that is parallel to the centerline of the shrub, but not directly under the shrub, as shown. Open the data file in the FV software.
International Contact Europe Contact Distributors. Instructions confusing or unclear.
Instructions confusing or unclear. A There are several considerations when determining if a canopy is too small. If you still have the install discyou can also partition your primary and install W XP on ane new partition from http: These procedures are referenced from various locations in this manual.
Les Dickinson Level 3 Expert Answers.
HP Scanjet 2200c and Windows 8
Masked rings are ignored when LAI is computed. I hope I have helped you Turn the HP Scanjet scanner so that the bottom is up. If quantum sensors are connected to port 1 or 2, the light sensor name and calibration information andd in this field as well.
Enter the remaining scattering inputs step 6 of Procedure 3.
The first decision to be made involves how you wish to deal with matching the sensors. It includes the file name, console firmware version, time stamp indicating when the file was created, and user remarks.
Plug the power cord into an electrical outlet. This command imports all A records or just the ones that are closest in time to the existing B records in the target file. Customers also shopped for.
LAI-2200C Plant Canopy Analyzer
A View caps can be used to prevent the sensor from seeing in a particular direction, allowing readings to be made near the edge of a plot, and reducing the total plot size necessary.
Ships from and sold by Omnitec Design. The email address entered is already associated to an account. The experienced application scientists and analysts at LI-COR are committed to providing the highest level of customer service and support.
LAIC | Making Basic Measurements
Save the file to retain any changes. At the A site, log a 4A or 3A sequence depending on the size of the clearing.
A Gap fraction data at different angles potentially hold two types of information: Or, if you assume an extinction coefficient leaf angle distributiona line quantum sensor can be used at one angle.
Try going to Hp solutions website and download the updated drivers for windows 7. Click OK again to close the dialog box. The data file for this shrub looks something like this: This can be done on the control unit described below or with the FV software see Procedure 3.
The ACF value is. Prepare to scan Open the scanner lid. Send an email to our authors to leave your feedback. Position the optical sensor above the canopy.